Error analysis in two-terminal impedance measurements with residual correction
PublisherIEEE-INST ELECTRICAL ELECTRONICS ENGINEERS INC
Rights accessOpen Access
Residual impedance correction in impedance analyzers when using an asymmetrical test fixture needs three reference measurements, usually open circuit, short circuit, and load (meaning an impedance close to the impedance under test). This paper provides an error estimate for impedance measurements that apply a simple open/short correction in spite of using an asymmetrical test fixture. Experimental results show that the minimal error is obtained for impedance values close to the geometric mean of the short-circuit and open-circuit impedances, and that the theoretical prediction is indeed an upper limit for the actual error.
CitationTorrents, J.M.; Pallas R. Error analysis in two-terminal impedance measurements with residual correction. IEEE Transactions on Instrumentation and Measurement. 2005, vol 54 núm 5, p. 2113-2116
Showing items related by title, author, creator and subject.
Sánchez Terrones, Benjamín; Praveen, A.; Bartolome, E.; Soundarapandian, K.; Bragós Bardia, Ramon (2014-06-01)
Restricted access - publisher's policyIn this paper, we present a framework to analyze the propagation of measurement errors through the body composition equations. This is done through the equations for body composition assessment (BCA), for single/multifrequency ...
Measurement errors in multifrequency bioelectrical impedance analyzers with and without impedance electrode mismatch. Bogónez Franco, Francisco; Nescolarde Selva, Lexa Digna; Bragós Bardia, Ramon; Rosell Ferrer, Francisco Javier; Yandiola, Iñigo (2009-06-03)
Restricted access - author's decisionThe purpose of this study is to compare measurement errors in two commercially available multi-frequency bioimpedance analyzers, a Xitron 4000B and an ImpediMed SFB7, including electrode impedance mismatch. The comparison ...
Torrents i Dolz, Josep Maria; Pallás Areny, Ramón (Institute of Electrical and Electronics Engineers, 2005-10)
ArticleResidual impedance correction in impedance analyzers when using an asymmetrical test fixture needs three reference measurements, usually open circuit, short circuit, and load (meaning impedance close to the impedance under ...